محصولات » OPTIKA » میکروسکوپ پلاریزان
This post is also available in:
Polarized light microscopy is an optical microscopy technique involving polarized light. Simple techniques include illumination of the sample with polarized light. Directly transmitted or incident light can, optionally, be blocked with a polariser orientated at 90 degrees to the illumination. These illumination techniques are most commonly used on birefringent samples where the polarized light interacts strongly with the sample and so generating contrast with the background. Polarized light microscopy is used extensively in optical mineralogy.
As polarised light passes through a birefringent sample, the phase difference between the fast and slow directions varies with the thickness, and wavelength of light used. The optical path difference (o.p.d.) is defined as
o . p . d . = Δ n x t
where t is the thickness of the sample.
This then leads to a phase difference between the light passing in the two vibration directions of
δ = 2 π ( Δ n x t / λ )
For example, if the optical path difference is λ / 2 , then the phase difference will be π , and so the polarisation will be perpendicular to the original, resulting in all of the light passing through the analyser for crossed polars. If the optical path difference is n x λ, then the phase difference will be 2 n x π , and so the polarisation will be parallel to the original. This means that no light will be able to pass through the analyser which it is now perpendicular to. The Michel-Levy Chart arises when polarised white light is passed through a birefringent sample. If the sample is of uniform thickness, then only one specific wavelength will meet the above condition described above, and be perpendicular to the direction of the analyser. This means that instead of polychromatic light being viewed at the analyser, one specific wavelength will have been removed. This information can be used in a number of ways:
– If the birefringence is known, then the thickness, t, of the sample can be determined
– If the thickness is known, then the birefringence of the sample can be determined
As the order of the optical path difference increases, then it is more likely that more wavelengths of light will be removed from the spectrum. This results in the appearance of the colour being “washed out”, and it becomes more difficult to determine the properties of the sample. This, however, only occurs when the sample is relatively thick when compared to the wavelength of light
هیچ دیدگاهی برای این محصول نوشته نشده است.
ذخیره نام، ایمیل و وبسایت من در مرورگر برای زمانی که دوباره دیدگاهی مینویسم.
لطفا پاسخ را به عدد انگلیسی وارد کنید:
دیدگاه شما *